MICRO-SEPAROMETER MARK V DELUXE OPERATION MANUAL
CLAY MONITOR TEST PROCEDURE
The DELUXE MICRO-SEPAROMETER can be used for the Clay Monitor Test in the following manner.
Prepare the DELUXE MICRO-SEPAROMETER as stated in section 7.0.
Momentarily depress the "ON" switch. The annunciator lamps located on switches A through G in the TEST
SELECT section will then be scanning for your selection. ' Depress switch "F" to initiate the standard automatic program
for Clay Monitor testing.
The annunciator lamps in the SYRINGE section will indicate that manual control can be executed on the syringe
drive mechanism. Depress the "UP" pushbutton. This will move the syringe mechanism to the upper limit. The
annunciator lamp in the PROGRAM section will indicate the turbidimeter can be utilized on demand. Depressing the
"READ" switch will turn on the turbidimeter and the adjust capability of the turbidimeter.
Insert sample into syringe and attach to syringe drive mechanism.
Momentarily depressing the "DOWN" pushbutton will cause the syringe drive to dispense the fuel sample in
approximately 15 seconds.
COPYRIGHT 1984 EMCEE ELECTRONICS, INC.
REVISION DATE: FEBRUARY 12, 1987