MICRO-SEPAROMETER MARK V DELUXE OPERATION MANUAL
When securely in place depress the "START" pushbutton. This will initiate the automatic program.
Meter Adjust Period #1
Meter Adjust Period #2
Syringe Drive Down
The automatic program starts with a read meter warning followed by a 10 second full scale adjustment period.
During this period adjust the display to 100 using the meter adjust up or down switches.
After the emulsification period, remove the syringe from the emulsifier and insert the plunger with the help of the
wire aid. Remove the plug from the bottom, replace it with an ALUMICEL® COALESCER marked DIESEL, and place the
whole assembly onto the syringe drive mechanism. Attach ground lead between ground jack on side of syringe drive and
ALUMICEL® COALESCER. Place a waste container beneath the syringe to collect the unwanted fuel.
At the end of the second meter adjust period the syringe will start down. Remove and empty the sample vial and
prepare for sample collection. The audio alert and the C/S indicator will indicate the proper time to collect the last 15
milliliters from the syringe.
Place the sample vial in the turbidimeter and rotate to the marked position.
When the steady tone starts the meter will automatically turn on. An additional one second tone during the meter
on period will indicate the proper time to read the meter. Record the reading and report as Micro-Separometer rating
COPYRIGHT 1984 EMCEE ELECTRONICS, INC.
REVISION DATE: FEBRUARY 12, 1987