MICRO-SEPAROMETER MARK V DELUXE OPERATION MANUAL
MINISTATIC TEST PROCEDURE
The Ministatic test utilizes the syringe drive mechanism. For this test it is necessary to move 50 ml. of fuel
through the syringe in 30 seconds.
Prepare the DELUXE MICRO-SEPAROMETER as stated in section 7.0.
Momentarily depress the "ON" switch. The annunciator lamps located on switches A through G in the TEST
SELECT section will then be scanning for your selection. Depress switch "G". This will initiate the MINISTATIC TEST
The annunciator lamps in the SYRINGE section will indicate that manual control can be executed on the syringe
drive mechanism. Depress the "UP" pushbutton. This will move the syringe mechanism to the upper limit. The
annunciator lamp in the PROGRAM section will indicate the turbidimeter can be utilized on demand. Depressing the
"READ" switch will turn on the turbidimeter and the adjust capability of the turbidimeter.
Insert sample into syringe and attach to syringe drive mechanism.
Momentarily depressing the "DOWN" pushbutton will cause the syringe drive to dispense the fuel sample in
approximately 30 seconds.
COPYRIGHT 1984 EMCEE ELECTRONICS, INC.
REVISION DATE: FEBRUARY 12, 1987