Under normal conditions, the test set will maintain
the factory alignment over a long period of time. It
is necessary to realign the test set if any component
except the batteries is changed. Alignment procedures
of Meter Zero
a Set the polarity switch to OFF.
b. Adjust the meter zero adjustment screw located
directly below the base of the pointer for a 0 meter
of Meter OHMS Position
a. Test Equipment Required. The only test equip-
ment required is a 100-ohm, 1 percent, -watt
(1) Disconnect the unit from the case by remov-
ing the four front panel screws.
(2) Set the polarity switch to FWD PNP.
(3) Set the. function switch to OHMS E-B.
meter pointer indicates full scale on the OHMS scale.