TM 11-6625-291-14/TO 33A1-7-23-1
2-10. MEASUREMENT OF OUTPUT POWER USING TS-585A/U. (CONT)
If the characteristic impedance is unknown, see paragraph
to find the characteristic impedance
of the circuit
Set meter multiplier knob to 100.
3. Output Meter
Set equipment under test on.
Set meter multiplier knob to setting where
meter reading is slightly to right of center
More accurate readings are taken when
meter needle points near center of
Read the power in milliwatts by: meter multiplier setting
times the meter reading. Read the power in decibels by:
meter multiplier setting plus the meter reading.