TM 11-6625-291-14/TO 33A1-7-23-1
2-11. MEASUREMENT OF OUTPUT POWER USING TS-585B/U, TS-585C/U, OR TS-585D/U.
This task covers:
Equipment under test off
The output meter is not fused. Do not connect the output
meter to equipment rated more than 5 watts.
Test leads (1) and
1. Output Meter
Connect test leads to binding posts.
binding posts (2)
If equipment under test is grounded, connect test lead from
binding post (marked G) to ground of equipment under test.
Impedance control (3)
If the characteristic impedance is unknown, see paragraph
to find the characteristic impedance
of the circuit