1 through 13.
in the vertical plane, any additional error
incurred will not be greater than one
a. General Use.
percent of full-scale. If the multimeter is
(1) When the approximate value of the quan-
to be permanently operated in the ver-
tity being measured is known, it is not
tical plane, it should be re-calibrated.
necessary to first use the multimeter's
search mode. Use the digital mode only,
but remember to resolve the meter
pointer proceeding from left to right
the Xl OHMS range in the search mode,
with the digit wheels.
the instrument will not indicate zero re-
sistance with the test lead terminals
rotation of the meter pointer in the
shorted together. The value indicated is
horizontal plane. Should it become occa-
sionally necessary to use the model 300M
and is nominally 0.1 ohm. When low