MODEL 101 INSTRUCTIONS
LINE SEPARATION ADAPTER
This adapter is used to divide the two conductor cord so that AC current measurements can be made on equipment with
a standard line cord.
It will increase sensitivity and ranges of the Model 310-Model 10 combination by 10 times and 20 times. Also the adapter
has provisions for voltmeter prods for voltage measurements.
Uses for Model 101 (With Model 310-10 combination V-O-M, clamp-on ammeter).
Fast checks on current of appliances, home freezers, radios, TV, etc.
Checks for shorts, etc. in home radios, TV sets and other equipment.
Checks on current of motors, starting and running.
Current Capacity of Model 101
Max. Capacity Amperes
Divide by 10
Divide by 20
For current readings on 30 amperes, on equipment having two conductor line cords.
(1) Plug line of cord appliance or apparatus to be measured into the "Direct" outlet of the Model 101 (Fig I) Plug model
101 into wall outlet.
(2) With Model 10 adapter and Model 310 V-O-M connected and set properly for measuring AC current, snap Model
10 through the hole In Model 101 and take readings, starting at the highest current range. (Fig 2). Reset range
switch on Model 10 for greatest pointer deflection.