24. Checking Unpacked Equipment
a. Inspect the equipment for damage during
As normally supplied by the factory, the unit need
shipment. If the equipment has been damaged,
only be placed on a horizontal flat surface in an
report the damage on DD Form 6.
upright position for use. If rack mounting is
b. Check the equipment against the component
necessary, the rack mounting kit furnished with the
listing in the operator's manual and the packing slip
instrument should be used. Figure 2-2 illustrates the
to see if the shipment is complete. Report all dis-
assembly of this kit.
crepancies in accordance with the instructions of TM
38-750. The equipment should be placed in service
even though a minor assembly or part that does not
The line voltage switch located on the rear
affect proper functioning is missing.
panel of the instrument must be set to the
c. Check to see whether the equipment has been
proper line voltage (115 or 230 volts). Verify
modified. Equipment which has been modified will
that the proper switch setting has been made
have the MWO number on the front panel, near the
before connecting equipment to the power
nomenclature plate. Check also to see whether all
currently applicable MWO have been applied.
The ME-202C/U is designed to be operated from a
(Current MWO applicable to the equipment are
single phase power source having the following
listed in DA Pam 310-7.)
Test Equipment, and Materials
a. Voltage of 115- or 230-volt nominal, 1O%.
Required for Installation
No special tools are needed for normal installation of
c. Power capability of 15 watts.